The QCM-I measuring unit is a high-sensitivity mass sensor, which measures the change in frequency of a quartz crystal resonator and as a label free biosensor it measures the “wet mass” of the adsorbed layer in processes occurring at or near the sensor surface. The measuring principle is based on impedance analysis of the quartz crystal. The resonant frequency and the bandwidth of the resonant conductance curve are determined. The bandwidth or full width at half maximum, ( FWHM ) is direct correlation with the quality factor ( Q ) which is by definition the inverse of the well known dissipation ( D ).
Control and measurement:
The BioSense software performs full control of the QCM-I instrument. Two measurement modes are available:
Calculation: various QCM and ad-layer parameters can be calculated using standard or custom made models. All these parameters can be displayed on the screen, printed, saved or can be exported for further evaluation.
Quartz crystal sensors with Au or with ITO electrodes and a variety of pre-coatings are available. The modular sensor holder allows using either 14 mm diameter or 1” diameter quartz crystals.
In conventional QCM, the electrodes are typically made of gold which is an ideal electrode material except when applying for electrochemical measurements or for optical investigations using inverse optical microscopes. On the QCM-ITO sensor, transparent ITO electrodes are used and hence providing a more stable electrochemical measurement and can be used for optical investigation as well. For sensor details see separate sensor data sheet.